Station Name, Device Serial Number, Manufacturer Model Name,Date of acquisition,Time of acquisition,Background calibration factor,RC sphere 1 3D ROI,RC sphere 2 3D ROI,RC sphere 3 3D ROI,RC sphere 4 3D ROI,RC sphere 5 3D ROI,RC sphere 6 3D ROI,RC sphere 1 MAX,RC sphere 2 MAX,RC sphere 3 MAX,RC sphere 4 MAX,RC sphere 5 MAX,RC sphere 6 MAX,RC sphere 1 peak,RC sphere 2 peak, RC sphere 3 peak, RC sphere 4 peak, RC sphere 5 peak, RC sphere 6 peak
ECAT,ECAT,ECAT,24012017,18:00:00, 1.00643, 0.961964, 0.967167, 0.924284, 1.01156, 1.01892, 0.681488, 1.33605, 1.38402, 1.23958, 1.39131, 1.46612, 1.02119, 0.992899, 0.941632, 0.933099, 0.971838, 0.845485, 0.442275
