Station Name, Device Serial Number, Manufacturer Model Name,Date of acquisition,Time of acquisition,Background calibration factor,RC sphere 1 3D ROI,RC sphere 2 3D ROI,RC sphere 3 3D ROI,RC sphere 4 3D ROI,RC sphere 5 3D ROI,RC sphere 6 3D ROI,RC sphere 1 MAX,RC sphere 2 MAX,RC sphere 3 MAX,RC sphere 4 MAX,RC sphere 5 MAX,RC sphere 6 MAX,RC sphere 1 peak,RC sphere 2 peak, RC sphere 3 peak, RC sphere 4 peak, RC sphere 5 peak, RC sphere 6 peak
ECAT,ECAT,ECAT,24012017,18:00:00, 0.998382, 0.909588, 0.880700, 0.878627, 0.733698, 0.696227, 0.440954, 1.32038, 1.26916, 1.33715, 1.05896, 1.02118, 0.636643, 0.990720, 0.969225, 0.928284, 0.736751, 0.607352, 0.321299
